Tip: For personalized troubleshooting assistance, contact us at firstname.lastname@example.org.
ELISA Troubleshooting: High Variation
High well-to-well variation can be a symptom of a flawed experiment, or it can be a simple error that’s easy to fix. Errors in plate handling or loading, or mistakes in the usage of reagents are all common causes of well variance in ELISA. Use this troubleshooting guide to help solve some of the common sources of error that drive high well-to-well variability.
|Plate handling problems||Plates stacked during incubation||Do not stack plates to ensure even temperature distribution|
|Plates warmed unevenly before use||Allow plate temperature to stabilize before use to avoid edge effects|
|Bubbles in wells||Check for bubbles before incubations|
|Uneven washing||Wash all wells uniformly|
|Check plate washer for obstructed ports|
|Reagent problems||Incomplete reagent mixing||Make sure all reagents are homogenous before use|
|Inconsistent storage||Make sure all samples and reagents are stored under uniform conditions|
Keywords:- elisa, troubleshooting, well-to-well, variation, high variation, intra-assay variation